|Product||Radar level meter|
|Measure range||0-20 meters|
|Process Connection||Thread, Flange|
|Process Pressure||Normal pressure|
|Signal Output||4-20mA (Two-wire/Four)|
|Power supply||DC(6~24V)/ Four-wire
DC 24V / Two-wire
Be installed in the diameter of the 1/4 or 1/6.
Note: The minimum distance from the tank
wall should be 200mm.
Note: ① datum
②The container center or axis of symmetry
The top conical tank level, can be installed at
the top of the tank is intermediate, can guarantee
the measurement to the conical bottom
A feed antenna to the vertical alignment surface.
If the surface is rough, stack angle must be used
to adjust the angle of cardan flange of the antenna
to the alignment surface.
(Due to the solid surface tilt will cause the echo attenuation, even Loss of signal.)